See! Diagnose! Cure!
Fluorograph (X-ray system for chest). Diagnostic X-ray system BReeZe EBRX 01, ARCOM (Italy)
Diagnostic x-ray system (fluorograph) BreeZe EBRX 01 (manufactured by Arcom International, S.R.L., Italy)
is designed to perform general radiography of patients in a standing position or sitting on a chair, hospital gurney or wheelchair.
The system is recommended for use in medical institutions or departments of pulmonology and phthisiatrics for chest X-ray (fluorography).
A fixed focal length of 150 cm and motorized up and down movement allows you to take the following shots (exposures):
thorax; abdomen; skull; spinal column; limbs.
Complete set of X-ray diagnostic system BreeZe EBRX 01:
1. Tripod EBRX 01
|The BreeZe X-ray system EBRX 01 consists of a column on which a non-rotating arm is fixed. |
A digital X-ray image detector and an X-ray emitter are located on the same axis of the console.
|X-ray collimator||Optica -20 (Varex USA) |
or R-302A (Ralco Italy)
|Focal length tube-detector||1500 mm|
|Electro-mechanical vertical movement of the arm with emitter and detector||1420 mm|
|Buttons for controlling the electric drive for moving the console up and down are located both in the emitter zone and in the detector zone|
|Lower position of the center of the radiograph from the floor||380 mm|
|Upper position of the center of the radiograph from the floor||1800 mm|
|Embedded X-ray raster||Option|
|Automatic exposure control device (AEC)||AEC ionization chambers (Varex USA)|
2. X-ray power supply
2. X-ray Power Device (high voltage generator) ODEL s.r.l. (Italy) Endeavor 40R R306.34, 40kW, including AEC (automatic exposure control function), membrane generator control console.
Due to the high conversion frequency, 400 kHz, it has stable output signal parameters.
|Power supply||380 V▒10%, 50▒1 Hz|
|Output power, kW||40 kW|
|Anode voltage range||40-150 kV|
|Anode current variation range||10-500 mA|
|Exposure time range||0.001 ÷ 6.3 s
|X-ray tube overload protection|
|Four programming modes available:|
2) kV, mA;
3) kV when working with the automatic exposure control function AEC;
4) Anatomy Mode
|AEC automatic exposure control function|
|Clinical dosimeter connection function|
|Displaying the value of the dose-area product on the generator control panel to determine the effective doses to patients|
|High voltage manual button|
At the request of the customer or technical specifications, it can be equipped with generators CMP200 40 kW, or generators with a capacitor storage CMP200EA 32 kW, 40 kW. It is also possible to complete with generators of other manufacturers similar in characteristics.
3. X-ray emitter X42 in C40 housing (IAE, Italy)
|Maximum tube voltage||130 kV|
|Heat capacity||150kJ (200 KHU)|
|Small focal spot||0.6 mm|
|Large focal spot||1.5 mm|
|Power at small focus||17 kW|
|Power at a large focus||47 kW|
More details >> X42 X-ray emitter in housing C40.
4. Detector Exprimer EVS-4343G, DRTECH
4. Flat-panel digital X-ray image detector Exprimer EVS-4343G (DRTECH, South Korea). p>
|Working field size||430 x 430 (17"x17")|
|Matrix Size||3072 x 3072 Pixel|
|Wired Data Interface||Gigabit Ethernet|
|Image acquisition time||No more than 4.5 seconds|
|Detector size||460 x 460 x 15 mm|
|Weight with battery||4.5 kg|
It is also possible to complete with flat-panel detectors of other manufacturers similar in characteristics and dimensions.
5. Automated workplace of X-ray laboratory assistant
6. Automated workplace of a radiologist
This system is completed Dose Area Product Measuring System VacuDAP and Ionization Chamber AEC Sensor, the use of which is mandatory according to Article 17 of the Law of Ukraine źOn the protection of humans from the impact of ionizing radiation╗.
Explanation of the information and advertising nature of the information posted on the site.
Dear visitors of our site. All information contained on the site is for informational and advertising purposes only. The information provided on the site is not exhaustive.Technical specifications may be changed or supplemented at any time.
This information cannot be used as official document, for example, for the preparation of the technical part of the tender documentation.
For more information, contact our experts. Contact Us >>